1.5T with the magnet in place
The external sound processor and headpiece are MR Unsafe and must be removed before entering a room containing an MR scanner.
A bandage must be applied using the bandaging protocol below.
Horizontal closed bore scanners with a static magnetic field of 1.5T.
Maximum spatial field gradient of 2,000 Gauss/cm (20 T/m).
Maximum MR system reported, whole body averaged specific absorption rate (SAR) of ≤ 2.0 W/kg at 1.5T
(Normal Operating Mode) for quadrature transmit RF body coils.
Maximum MR system reported, head averaged SAR of ≤ 3.2 W/kg at 1.5T (Normal Operating Mode) for quadrature transmit RF head coils.
RMS gradient field of 30 T/s and peak gradient field of 150 T/s.
Under the scan conditions defined above the HiRes Ultra implant is expected to produce a maximum temperature rise of <3°C after 15 minutes of continuous 1.5T scanning.
In MRI testing of unilateral and bilateral recipient conditions, respectively, the image artifact caused by the device extends from the HiRes Ultra implant approximately 8 cm using a gradient echo pulse sequence and >9.5 cm using a spin echo or gradient echo pulse sequence in a 1.5T scanner with the magnet in place. These artifacts may result in a loss of diagnostic information in the implant vicinity.
The recommended minimum duration of time post implant surgery prior to undergoing an MRI scan is 2 to 4 weeks in order to allow any inflammation to subside.
An MRI scan is not recommended if the patient has a fever.
Please note that the MRI Antenna Coil Cover, CI-7521, and bandaging supplies must be on hand at the time of the MRI procedure. Please contact Advanced Bionics Customer Service at email@example.com prior to the MRI procedure to request the MRI Antenna Coil Cover.